SW-102 UV 125-525MHz
V.S.W.R. meter Introduction
VSWR. Forward and reflected power direct digital readout, without any calibration.
Maximum measurable power range up to 120W.
low insertion loss (0.3 decibels or less) structure allows it to be permanently connected.
Build in Frequency counter (SW102UV only) .
All Function display ( SWR,FW,RW,Battery Level)
Feature & Specification
Frequency range: 125 - 525MHz
Measurable power range: 0.5 - 120W(+-0.5%)
Measurable S.W.R range: 1.00:1~19.99:1
Input read frequency counter (must be < 0.5 W)
Maximum power: 120W
Size: W 76 X H 77 X D35 mm.
Weight: < 220 grams
Battery : 3.7V Li-ion Rechargeable Battery
* Not for the digital radio.
Interface: N-J / SO239 Interface
Interface: N-J Interface/ SO239
MAX. Power in :120 W (Measurable power,Radio must be ANALOG)
V.S.W.R1.00 :1 ~ 19.99:1
Frequency Range COVER : 125MHz~525MHz SW102UV
Insertion Loss< 0.15dB(VHF) , < 0.25dB(UHF)
Frequency counter : SW102UV only ( TX input must be minimum 0.5W + - 0.5%)
Battery Level Display : Yes
Battery Build-in :3.7V Li-ion Battery
Auto Power Off : 1-9 mins
Size: 66*68*37 (mm)
Charger in 5V (micro usb)
Back-light AUTO OFF
Interface (in and out )N / SO239 (SL16)
Weight: 220g (include battery)
Press [Red button] 2 second to power on or power off
MENU Display (Note: Version for up to V3.8)
Mode No.Item Adj
1 Adjustment Forward Power -99 to +99 Fine Tune Level %
2 Adjustment Reverse Power -99 to +99 Fine Tune Level %
3 Adjustment Time Base -99 to +99 Frequency Counter ,O.S.C. Time Base
4 LCD DIM (Back Light Time)0-9 , "0" is always on, 1 min to 9 mins
5 POWER OFF 0-9 "0" is always on, 1 min to 9 mins
* Supplier, the product will add functionality without having to give notice
Set & Save:
Press MENU [Blue button]
Press F1 [Red button)(Select 1-5 Function)
Press [Blue button] ( - ), [Yellow button] (+)
Press the [Red button ]for 2 seconds
Press the [Blue button] to SAVE and EXIT
How to check the S.W.R. of your Antenna :
Connect to "TX" on Radio output and Antenna to "ANT". Proper adapter /cable is required if your device has different type of connectors.
Measurement V.S.W.R. : (see Pic.1)
SW-102 TX connect Radio RF out
Radio TX , SW-102 with display S.W.R. measure data.
Do not affect the test in the following cases, there are objects nearby, placed horizontally.
How to check the transceiver RF output Watt :
Cable Connect to "TX" on Radio output and dummy Load to "ANT". Proper adapter /cable is required if your device has different type of connectors.
Measurement Power: (see Pic.2)
SW-102 TX connect Radio RF out
SW-102 ANT connect 50 ohm Dummy Load
Push Radio TX , SW-102 display is Power measure data.
(SW-102 is not for D.M.R. system)
How to display rotate : (up to Ver. 2.02 only )
Press Yellow key to rotate display in 180 degree ,
Press Yellow key to normal .
1x SURECOM SW-102-UV Power & S.W.R. Meter (N - connector)
1x Power PSU 100-240V
1x Charger Cable
1x English Users Manual
Version information of date:
VER.3.3~V4.5 (P.C.B. 001) Production on 2016-2017
VER.5.1 (P.C.B. 001) Production on 2017-2018
VER.2.0 ~V2.03 (P.C.B. 002) Production on 2018-2020
VER.2.06 ~2.08 (P.C.B. 002) Production on 2020-2021
VER,2.4 ~V2.7 (P.C.B. 003) Production on 2021- 2023
The first from :
P.C.B. 001 (Old version ) V3.3,V3.8,V4.3, V4.5 can up to V5.1 .
P.C.B. 002 V2.02,V2.03,V2.06 ,V2.07 ,V2.08 (Version V2.0 to V2.06 (is same V2.07 V2.08)..... , the watt meter refenece is best than V3.3,V3.8,V4.3, V4.5,V5.1 .)
P.C.B. 003 V2.4 ~ V2.7 (New PCB design for S.W.R.) .
(* V2.7 will be cancelled the efficiency % ,sorry for the inconvenience)
If there is any change in the factory, there will be no further notice.
N - CONNECTOR
SO239 (SL-16)- CONNECTOR